Generally, supplier of commercialized contact angle meter will suggest you to choose surface tensiometer and use Washburn method to measure water adsorbing of fiber. But, in practical application, due to the updated speed of balance (Sartorius's balance is 12 data/s and MTs is 92data/s.) and the result of contact angle by Washburn method has no comparability with the result from the Optical contact angle meter. So, KINO will suggest you choose optical contact angle meter with high speed camera (400-2000FPS) to measure drop adsorbing and contact angle VS time.
Due to surface roughness, chemical diversity and heterogeneity of the surface of fiber, we will not use contact angle goniometer with circle, ellipse or tangent method, or Young-Laplace Equation fitting method, because circle ftting, ellipse fitting and tangent method is just a geometric value of goniometer and has nothing to do with the surface chemistry. And the measured value by these method has no repeatability and Comparability. So, only contact angle meter with Young-Laplace equation fitting method is real contact angle meter. But the biggest flaw of Young-Laplace equation fitting method and ADSA-P is that it cannot be used to measure contact angle of nonaxisymmetric drop shape. Now, only ADSA-RealDrop method can measure nonaxisymmetrical drop shape and calculate the intrinsic contact angle.
We used optical contact angle meter model SL200KS and was equipped with high speed camera with speed 162-3400FPS to measure drop adsorbing and dynamic contact angle vs time. And we recognized the difference between several samples by the adsorbing curve and Initial Contact Angle.
|Initial Contact Angle|
1, of the sample blank
2, of the sample 2
3, of the sample 5
4, of the sample 10
5, of the sample 15
6, of the sample 20
Hot keywords of USA KINO:contact angle, contact angle measurement, contact angle meter, contact angle goniometer, surface tensiometer, interfacial tensiometer, surface tension measurement, surface tension, surface tensiometry, contact angle measurement equipment and device, calculating surfac free energy, Determining Critical Micelle Concentration (CMC) of surfactant, made in China Method for choosing surface tensiometer NEW Method for choosing contact angle meter (goniometer) NEW Method for choosing interfacial tension meter NEW
MainPro : contact angle,contact angle meter,contact angle goniometer,surface tension,surface tensiometer